Replica tape for measuring blasted surface profiles. By placing the Tqc replica tape on the surface and rubbing over it, the Rt (total roughness) or peak-to-valley height of the profile can be taken and then measured with a film thickness meter.
1. TQC Replica tape consists of a layer of compressible foam affixed to an incompressible polyester substrate.
2.When pressed against a roughened/gritblasted (steel) surface, the foam collapses acquiring an impression of the surface.
3.Placing the compressed tape between the anvils of the SP1570 micrometric thickness gage, and subtracting the contribution of the incompressible substrate (50 micrometers or 0.002 inches), gives a measure of the surface profile.
One box of Testex contains tape for 50 measurements.
Type Testex-tape Medium, foam layer thickness 0.4 mills/10µm not to be used with a film thickness meter
Type: Testex-tape coarse Minus, range 0.5 to 1.0 mils/ 12 to 25 µm
Type: Testex-tape COARSE range: 0.8 to 2.5 mils / 20 to 64 µm
Type: Testex-tape X-COARSE range:1.5 to 4.5 mils / 38 to 115 µm
Type: Testex-tape X-coarse Plus, range 4.6 to 5.0 mils / 116 to 127 µm